High-end microscope at a price of the middle class one. Optical performance and usability of these microscopes allow SmartOPTIC successfully compete even with excellent German brand microscopes with a long tradition. It is a device often and willingly chosen by dentists in Poland.

The sum of the functional characteristics of the SmartOPTIC microscope gives a product that many call "optimal" having regard to ratio of quality to the price.

The microscope SmartOPTIC modular design allows any configuration suited to the expectations of the operator.
By default, the microscope is available in two basic configurations:

• SmartOPTIC BASIC with fixed binocular 45
• SmartOPTIC ERGO with inclinable binocular in the range 0-195

Below the summary of the most important features of the microscope and its technical details.

## Binoculars

Equipped with a wide angle oculars with independently adjustable diopters for each eye. Flexible eye sockets allowthe user  to work in your own glasses. Depending on the version of the microscope the binocular can be  fixed or inclinable.

 SmartOPTIC BASIC: fixed binocular 45° SmartOPTIC ERGO: inclinalbe binocular 0-195°

## High-quality optics, five degrees of magnification

The applied optics provides a full range of magnification: from the smallest (x1, 3) to the largest (x 27) depending on the configuration.
See Table of magnifications.

## LED light source as a standard

The light source is a modern80W LED , which gives a very bright, white light. Its' service life is planned for 50 000 hours.

## 3 years guarantee

The reliability confirmed with the 3 years guarantee.

## A wide selection of accessories

SmartOPTIC microscope can be fitted with a range of accessories, such as inclinable binocular( 0-195 degrees), the arm extension, binocular extender, etc.

See more: Accessories

## Accurate focus setting

Built-in lens fine focus knob allows precise focusing. Essential when using high magnifications.

## Many options of mounting

Available versions: wall, ceiling, floor, table and mobile on wheels.

## Various types of vision tracks

Possibility of video track installation for DSLR cameras, SLR cameras, dedicated cameras and popular Handycam.

See more: Imaging

## Available filters

Depending on needs, the microscope may be equipped with a filters: orange, green, or polarization.

## Compact structure, aesthetic and modern design

Design gives complete freedom of movement, even in small offices.

 Basic configuration of the SmartOPTIC microscope consists the following elements:   Binocular: fixed angle of 45º; wide-angle; oculars with independent diopter adjustment; Objective lens: 250mm Light source: LED; efficient fiber optic cable; Filter: orange or green; Arm length: extended (90cm); Detailed information on the individual components can be found under Accessories

 The modular design allows the microscope to equip it with various components depending on your needs.   Binoculars: fixed angle of 45º; fixed angle of 90º; with adjustable tilt angle of 0-195º; Objective lenses: 200mm; 250mm; 300mm; 400mm; Lightning system: halogen 3000K 37klux; LED 6000K 50klux; Xenon 6000K 105klux; efficient optical fiber; Filters: green; orange; laser filter; polarizing filter; Mounting options: ceiling; wall; floor; table; mobile on wheels; Arm of the microscope: standard 60cm; extended 90cm; additional extension of 28cm; Imaging: dedicated camera with beamsplitter; single beamsplitter; double beamsplitter; adapter for DSLR camera; adapter for camcorder type of camera"; adapter for Sony Nex camera; adapter for CCD camera; LCD mast; Equipment: binocular extender; rotation ring; LCD mast; shelf for printer; autoclavable handle covers; Detailed information on the individual components can be found under Accessories

Wall

Ceiling

Mobile on wheels

Table

Magnifications Table for the SmartOPTIC microscope . The table takes into account the magnification of the microscope for various configurations and installed components: lens (200mm, 250mm, 300mm, 400mm), oculars (x10, x12, 5), binocular (inclined- F135, inclinable- F170).

 objective: 200mm 250mm 300mm 400mm binocular: inlined inclinable inlined inclinable inlined inclinable inlined inclinable ocular knob position f135 f170 f135 f170 f135 f170 f135 f170
 10x/16mm 0,4 2,70 3,40 2,16 2,72 1,80 2,27 1,35 1,70 10x/16mm 0,6 4,05 5,10 3,24 4,08 2,70 3,40 2,02 2,55 10x/16mm 1,0 6,75 8,50 5,40 6,80 4,50 5,66 3,37 4,25 10x/16mm 1,6 10,80 13,60 8,64 10,88 7,20 9,06 5,40 6,80 10x/16mm 2,5 16,88 21,25 13,50 17,00 11,25 14,17 8,42 10,63
 12,5x/16mm 0,4 3,37 4,25 2,70 3,40 2,25 2,83 1,69 2,13 12,5x/16mm 0,6 5,06 6,38 4,05 5,10 3,38 4,25 2,53 3,20 12,5x/16mm 1,0 8,44 10,63 6,75 8,50 5,63 7,08 4,20 5,30 12,5x/16mm 1,6 13,50 17,00 10,80 13,60 9,00 11,32 6,74 8,50 12,5x/16mm 2,5 21,08 26,58 16,88 21,25 14,08 17,71 10,54 13,33

TOP